You are presented with a paragraph from a COVID-19 literature. Classify the topics of this piece of text by choosing from the following categories:

- Epidemic Forecasting: The content relates to predictions about the spread of COVID-19.
- Treatment: The paragraph discusses therapeutic methods or medication for COVID-19.
- Prevention: The content covers measures to prevent the spread of COVID-19.
- Mechanism: The paragraph explains the biological mechanisms of the virus.
- Case Report: The content is a report of a COVID-19 case.
- Transmission: The paragraph discusses how COVID-19 is transmitted.
- Diagnosis: The content is about diagnosing COVID-19 in individuals.

Your classification should be provided as an element within a JSON array. The array should be formatted with topics enclosed in double quotes and separated by commas, following the JSON format, for example: ["Topic1", "Topic2", ...]. Include all applicable topics within the array. Your output should be a valid JSON array, and no additional text should be included in your response.

Article:

BACKGROUND: To conserve personal protective equipment (PPE) and reduce exposure to potentially infected COVID-19 patients, several Californian facilities independently implemented a method of acquiring portable chest radiographs through glass barriers that was originally developed by the University of Washington. METHODS: This work quantifies the transmission of radiation through a glass barrier using six radiographic systems at five facilities. Patient entrance air kerma (EAK) and effective dose were estimated both with and without the glass barrier. Beam penetrability and resulting exposure index (EI) and deviation index (DI) were measured and used to adjust the tube current-time product (mAs) for glass barriers. Because of beam hardening, the contrast-to-noise ratio (CNR) was measured with image quality phantoms to ensure diagnostic integrity. Finally, scatter surveys were performed to assess staff radiation exposure both inside and outside the exam room. RESULTS: The glass barriers attenuated a mean of 61% of the normal X-ray beams. When the mAs was increased to match EI values, there was no discernible degradation of image quality as determined by the CNR. This was corroborated with subjective assessments of image quality by chest radiologists. The glass-hardened beams acted as a filter for low energy X-rays, and some facilities observed slight changes in patient effective doses. There was scattering from both the phantoms and the glass barriers within the room. CONCLUSIONS: Glass barriers require an approximate 2.5 times increase in beam intensity, with all other technique factors held constant. Further refinements are necessary for increased source-to-image distance and beam quality in order to adequately match EI values. This does not result in a significant increase in the radiation dose delivered to the patient. The use of lead aprons, mobile shields, and increased distance from scattering sources should be employed where practicable in order to keep staff radiation doses as low as reasonably achievable.
["Prevention"]