This type high purity hydrogen generator is a new type product developed on the foundation of the technology of astronautic fuel cells. The working principle of hydrogen generator is, hydrogen is produced by the electrolysis of water, and the produced oxygen is released to the atmosphere. It is made up of these systems such as the electrolyzing and separating pool, switch power, pressure controller, desiccation and purification and flow display.
The core part of the instruments is electrolyzing and separating pool, in which the positive pole is of tube type structure, the negative pole is of sheet metal shape, and both the poles are made of stainless steel. The electrolyzing &separating pools can store electrolyte, produce hydrogen as well as release hydrogen at the same time, and it is the best replacement to sheet shaped electrolyzing pools. The instruments have advantage such as the large electrolyzing area, low pool temperature, large production of hydrogen with high purity. The instruments are specially installed with the devices that can prevent liquid from returning, so the instruments can be effectively guaranteed that the phenomenon of liquid returning doesn't occur during working.
2. Main Technical Parameter
|Hydrogen purity||99.99% (relative oxygen content)|
|Max. Output flow||0-2000 ml /min|
|Max. Output pressure
||0.4Mpa (special pressure, please tell us)
|Working condition||220V/110V, 50HZ/60Hz|
|Environmental condition||Temperature: 0-40°C, Relative, humidity<85%|
|Without a lot of dust and corrosive gases|
|Maximum power||500 W|
|size||512 × 240 × 450 mm (L × W × H)|
English Manual Download
The maximum output flow of the gas generator is 2000ml/min, so please control your equipment's gas consumption below 1000ml/min.
One year warranty!
3. Characteristics of the Instruments
(1) It is so simple to operate the instruments that hydrogen can be produced just after the switch is opened. The pressure of released hydrogen is steady, and the flow is digitally displayed with LED, which is more ocular and striking.
(2) In ordinary using, the instruments just need to be supplemented with distilled water. The instruments can be used in succession or disconnectedly.
(3) The liquid storing, hydrogen electrolyzing and desorbing are integrated together in the instruments with the property of large electrolyzing area, low temperature of the pool, long service life, large amount and high purity of produced hydrogen.
(4) The instruments are specially installed with the devices that can prevent liquid from returning, so the instruments can be effectively guaranteed that the phenomenon of liquid returning doesn’t occur during operation. As a result, the discolor silica gel is not often to be changed.
(5) The adoption of the advanced switch powers can improve the electrolytic efficiency.
(6) The instruments substitute the high-pressure steel bottles, and make the laboratory instrument.