ASTM International - ASTM E537-20
Standard Test Method for Thermal Stability of Chemicals by Differential Scanning Calorimetry
|Publication Date:||1 February 2020|
|ICS Code (Physics. Chemistry):||07.030|
significance And Use:
5.1 This test method is useful in detecting potentially hazardous reactions including those from volatile chemicals and in estimating the temperatures at which these reactions occur and their... View More
5.1 This test method is useful in detecting potentially hazardous reactions including those from volatile chemicals and in estimating the temperatures at which these reactions occur and their enthalpies (heats). This test method is recommended as an early test for detecting the thermal hazards of an uncharacterized chemical substance or mixture (see Section 8).
5.2 The magnitude of the change of enthalpy may not necessarily denote the relative hazard in a particular application. For example, certain exothermic reactions are often accompanied by gas evolution that increases the potential hazard. Alternatively, the extent of energy release for certain exothermic reactions may differ widely with the extent of confinement of volatile products. Thus, the presence of an exotherm and its approximate temperature are the most significant criteria in this test method (see Section 3 and Fig. 1).
5.3 When volatile substances are being studied, it is important to perform this test with a confining pressurized atmosphere so that changes of enthalpy that can occur above normal boiling or sublimation points may be detected. As an example, an absolute pressure of 1.14 MPa (150 psig) will generally elevate the boiling point of a volatile organic substance 100 °C. Under these conditions exothermic decomposition is often observed.
5.4 For some substances the rate of enthalpy change during an exothermic reaction may be small at normal atmospheric pressure, making an assessment of the temperature of instability difficult. Generally, a repeated analysis at an elevated pressure will improve the assessment by increasing the rate of change of enthalpy.
Note 1: The choice of pressure may sometimes be estimated by the pressure of the application to which the material is exposed.
5.5 The four significant criteria of this test method are: the detection of a change of enthalpy; the approximate temperature at which the event occurs; the estimation of its enthalpy and the observance of effects due to the cell atmosphere and pressure.View Less
1.1 This test method describes the ascertainment of the presence of enthalpic changes in a test specimen, using minimum quantities of material, approximates the temperature at which these enthalpic changes occur and determines their enthalpies (heats) using differential scanning calorimetry or pressure differential scanning calorimetry.
1.2 This test method may be performed on solids, liquids, or slurries.
1.3 This test method may be performed in an inert or a reactive atmosphere with an absolute pressure range from 100 Pa through 7 MPa and over a temperature range from 300 K to 800 K (27 °C to 527 °C).
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use. Specific safety precautions are given in Section 8.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.