Analysis of Non-Metals in Sodium: Experience at The Reactor Research Centre, Kalpakkam
Measurement and control of non-metallic impurities like oxygen, hydrogen and carbon in sodium is of considerable importance in the successful long-term operation of the high temperature sodium systems of the fast reactors. In order to minimise corrosion of the structural materials, oxygen is to be maintained below 10 ppm level. Hydrogen estimation is important especially for the secondary sodium system of the reactor in order to monitor water to sodium leak in the steam generator. Control of carbon level is necessary to control carburisation/decarburisation of the structural materials. Considerable development has taken place during the last three decades in order to evolve suitable procedures for the estimation of these impurities in sodium. As a result, the laboratory methods developed in various laboratories have undergone various modifications to extend the sensitivities down to 1 ppm level. Foil equilibration methods such as vanadium wire equilibration, steel foil equilibration, and scandium foil equilibration have been developed in various laboratories to estimate oxygen, carbon and hydrogen respectively to ppb level. More recent development has been the electrochemical and diffusion type meters for on-line monitoring of these impurities in sodium. The analytical development work was started in our laboratory few years back. Several methods have been tried in keeping with the developments in this field. The evolution of the present methods have gone through several stages of modifications in order to evolve accurate, simple and fast procedures for the estimation of these impurities in sodium. This paper describes our experience in this field leading to the presently standardised procedures which are used routinely in our laboratory for the analysis of commercial sodium as well as reactor grade sodium.
KeywordsStainless Steel Vessel Commercial Sodium Hydride Hydrogen Sodium Loop Molecular Sieve Trap
Unable to display preview. Download preview PDF.
- 1.N.P. Bhat et al, The Determination of Oxygen in sodium; BARC-353, Bhabha Atomic Research Centre (1968).Google Scholar
- 3.J. Sannier and R. Vingot, Sodium sampling device on a circuit for impurity analysis, CEA-2369 (1963).Google Scholar
- 4.S.A. Meacham and E.F. Hill, Evaluation of the proposed ASTM Method for measuring oxygen in sodium utilising a Round Robin Programme, APDA-238 (1970).Google Scholar
- 5.R.F. Keough, Pacific North-west Laboratory Manual of Procedures for the analysis of metallic sodium, BNWL-MA-76 (1970).Google Scholar
- 6.R.D. Kale et al, Experience in the Operation of sodium purification loop in the Reactor Engineering Laboratory, RRC-11, Reactor Research Centre (1976).Google Scholar
- 7.J.A.J. Walker et al, Determination of Oxygen in sodium by distillation; TRG Report 952 (C) UKAEA (1963).Google Scholar
- 8.H.U. Borgstedt et al, Sodium distillation set up “NADESTAN-4” for analytical estimation, KfK Report 1941 (1973) Nuclear Research Centre, Karlsruhe.Google Scholar
- 10.J.D. Noden, A General equation for the solubility of Oxygen in liquid sodium; J. Brit. Nucl. Energy Soc. 12 (1973) 57.Google Scholar
- 11.S.A. Meacham and E.F. Hill, The Determination of Hydrogen in sodium, APDA-183, Atomic Power Development Associates (1966).Google Scholar
- 12.S.A. Meacham et al, The Solubility of Hydrogen in sodium, APDA-241, Atomic Power Development Associates (1970).Google Scholar
- 13.J.F.M. Rohde, Determination of total carbon content of sodium, SR-TN-7009-32, TNO, Holland (1970).Google Scholar