The Line EMI Meter measures electromagnetic interference in a single-phase AC power line. It plugs into a wall outlet (mains socket) and requires no battery. It immediately displays total line noise in millivolts peak to peak (range 1999.9 mVp-p, resolution 0.1 mVp-p) in the frequency range 10 KHz – 10 MHz (the frequency limits at which sensitivity is ½ of the maximum sensitivity). A speaker plays the sound of the EMI, amplitude demodulated, in order to identify the EMI source (such as an AM or shortwave radio station vs. a motor vs. an electric arc).
SPECIFICATIONS: Power Line Meter Model PLM
|Operating Voltage:||85-300 VAC 50/60 Hz.|
|Voltage Reading:||85-250 VAC; Accuracy +/- 1 VAC|
|Line Noise Range:||0.1-1999.9 mVp-p AC (differential)|
|Line Noise Accuracy:||+/- 8% @ 20 kHZ to 2 MHz; within a factor of 2 from 10 kHz to 10 MHz|
|Size:||5.2 x 3.6 x 1.6 inches; 132 x 91 x 41 mm|
|Power Cable:||Detachable 2-wire cord (appropriate to country)|