What are the differences between pulse tube and Gifford-McMahon cryocoolers?
To begin with, three differences have to do with price, intended orientation, and maintenance interval:  GM cryocoolers are generally less expensive than PT cryocoolers of similar cooling power. PT cryocoolers only function properly when they are operated in a purely vertical orientation with the 2nd stage pointing down. By contrast, GM cryocoolers can operate in any orientation; and even though cooling power is slightly reduced when the cold head is not operated vertically, the base temperature is normally unaffected. Most GM cryocoolers require scheduled maintenance every 10,000 operating hours, compared with 20,000 hours for PT cryocoolers. But theres also a fourth major difference, which has to do with vibration levels. Both GM and PT cryocoolers are mechanical refrigerators that produce vibrations, both at the cold heads room temperature mounting flange (which provides the interface for a vacuum shroud or vacuum chamber) and at the 2nd stage (where the sample being cooled is typically mounted). Which type of cryocooler to choose depends on the tolerable levels of vibration in your application or experiment and a couple of other factors:  A PT system is often a better choice if  Vibrations at the sample must be less than 20 m. Vibrations from the cold head need to be isolated from other equipment that interfaces with the cryostat. A GM system may be a better choice if  Vibrations on the order of 20 m at the sample are acceptable. The cryostat will not be rigidly mounted to any equipment that is sensitive to mechanical vibrations. A base temperature of 4 K or below is necessary but a PT system is beyond your budget limitations. The cold head will not be oriented vertically with the 2nd stage pointing down. For more information about these differences and what contributes to vibration, see this longer article on Janis cryocooler technology. Talk to one of our Janis engineers today.